Keysight Technologies (formerly Agilent) E3633A
160/200Watts of power
Single output, dual range
Front and rear panel terminals
Remote sense, OVP and OCP capability
16-bit programming resolution and high accuracy.
*The input power to the unit can be changed by removing the fuse holder
assembly (with a flat blade screwdriver), installing the proper fuse for the
desired input range (see below for fuse info), and then removing, rotating the
power-line input selector, then reinstalling it into the unit.
AC Input Ratings (selectable via rear panel selector)
115 Vac +/-10%, 47 to 63 Hz
230 Vac +/-10%, 47 to 63 Hz
0E9 100 Vac +/-10%, 47 to 63 Hz
*For 100 or 115 Vac operation, the correct fuse is 6.3 AT (Keysight part
2110-1030) and for 230 Vac operation, the correct fuse is 3.15 AT (Keysight
part number 2110-1031).